This type of transmitter can be used for open or vented to atmospheric type tank applications.
Following is calibration data for level transmitter of Direct mount and single system.
We have following information for LT Transmitter and diagram showing process
Density = 900 kg/m³
H – L = Nozzle Distance(mm)= 7600 mm
Capillary Density = 1090kg/m³
Pressure = ATM kgf/cm²G
Nozzle Low (mm) = 200 mm
Nozzle High (mm) = 7800 mm
LT Location(Below Low Nozzle)= 0
Following formula can be used to calculate LRV and URV
Low range value = LT Location * Capillary Density / 1000
Upper range value = (Nozzle Distance * Fluid Density / 1000) + (LT Location * Capillary Density / 1000)
To Calibrate properly please use the following ranges for Direct mount and single seal system Level transmitter.
LRV = 0% = 0 mmH2O
URV = 100% =6840 mmH2O
SPAN = 6840 mmH2O